The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||8 April 2004|
|PDF File Size:||6.83 Mb|
|ePub File Size:||19.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sending a text message to the display overrides the display state as set by the DISP command. Symmetry represents the amount of time per cycle that the ramp wave is rising assuming that the waveform polarity is not inverted.
You can issue triggers for sweeps or bursts using internal triggering, external triggering, or manual triggering. Invert the waveform relative to the offset voltage. Chapter 8 Specifications Specifications The variation in frequency of the modulated 33220a from the carrier frequency is called the frequency deviation. Table of Contents Add to my manuals Add. Used only when the Internal modulation source is selected AM: Errors are cleared as you read them.
The Agilent A assumes that all IP addresses and other dot- notation addresses are expressed as decimal byte values, and strips all leading zeros from these byte values. This generic message indicates that the desired edge time is limited to an upper or lower boundary which is dictated by the instrument hardware. You can also control the phase offset from the front panel or over the remote interface.
Agilent Technologies 33220A User Manual
Instructions and mounting hardware are included with each rack-mounting kit. Numeric Parameters Commands that require numeric parameters will accept all commonly used decimal representations of numbers including optional signs, decimal points, and scientific notation.
Select linear or logarithmic spacing for the sweep. Press any of the front-panel function keys except. Single, External, or Internal Carrier Waveforms: The function generator enables one burst mode at a time.
The isolation between chassis ground circuits and floating circuits is controlled by an optical isolation barrier and a serial link. At this point, the function generator outputs a continuous sweep from 50 Hz to 5 kHz if the output is enabled.
Quick Start One of the first things you will want to do with your function generator is to become acquainted with the front panel. A fifth storage location automatically holds the power-down configuration of the instrument. Another way to set the limits of a signal is to specify its high-level maximum and low-level minimum values.
The rear-panel 10 MHz In and 10 MHz Out connectors present only if Option is installed allow synchronization between multiple Agilent A function generators see connection diagram below or to an external 10 MHz clock signal. IP address to a device on a network.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
This rear-panel connector is used in the following modes: It covers both front panel and remote interface operation. If you change to a function whose manuual frequency is less than that of the current function, the frequency is adjusted to the maximum value for the new function. For a logarithmic sweep, the function generator varies the output frequency in a logarithmic fashion. Press the IP Setup softkey. Tutorial In order to achieve the best performance from the Agilent A, it may be helpful for you to gain a better understanding of the internal operations of the instrument.
The interface will self configure. The default is NORM. The edge time is manyal incremented. In FM, the frequency of the carrier is varied by the instantaneous voltage of the modulating waveform. Press the Hop Freq softkey and then set the value to Hz using the numeric keypad or the knob and cursor keys.
Chapter 5 Error Messages Execution Errors Settings conflict; edge time decreased due to pulse width For a pulse waveform, the function generator will automatically adjust the waveform parameters in the manial order as needed to generate a valid pulse: The default offset is 0 volts for all functions. Select the agient order for binary transfers in the block mode using the DATA: Burst Mode Chapter 3 Features and Functions Burst Mode Burst Mode You can configure the function generator to output a waveform with a specified number of cycles, called a burst.
Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 3322a0 ns for periods up to 10 seconds.
The function generator begins at the start frequency, sweeps to the stop frequency, and then resets back to the start frequency. In some cases, the amplitude limits are determined by the output units selected. If you want to modify the example programs, or write your own programs and compile them, you will need to install the Agilent IO Libraries Suite software.